Water Level Sensor For Deep Wells is made by building in high stable and reliable piezo-restive pressure sensor and high accurate circuit board into the stainless steel housing. Integrated construction and standard signal provide the user easy and convenient application in the local working place. The special cable connects with housing, can be immersed into the media for a long time.It has compact size, light weight and good stability; it can be used for water or liquid measure and control of medicine, metallurgy, electricity, mine, city water supply and drainage and hydrology etc.
a. Piezo-resistive Silicon Chip Employed
b. MEMS Technology
c. Cable Ventilated
d. Excellent long term stability
e. High accuracy and reliability
f. Completely submersible pressure transducer and cable
g. Compact, rugged design for easy installation
h. Minimal maintenance and care
i. Water Level Sensor For Deep Wells compatible with most monitoring equipment
j. Vented cable for automatic barometric compensation
Signal wire can use the double strand. We recommend the shielded wire in the situation with serious electromagnetic interference and please properly connect the wire to the ground .Don’t put the signal cables into a metal tube with other power cables or put them in the same slot, and decorate the wire far away from strong electric facilities.
Signal lines can be floating or grounding at any point in the signal circuit , the transmitter
housings can also be earthed or grounded.
As the transmitters ground through capacitive coupling, when checking insulation resistance, above 100V magnetometer can't be used , when checking circuit please under its max voltage,45V.
a. Please follow the diagram wiring to connect the circuit correctly. If connect backwards,due to the protection measures are adopted in transmitter, no signal output.
Transmitter will work the moment connecting the power, but the output signal is more stable and reliable after 30 minutes warming up after connecting.
b. The transmitter used in the medium which are compatible with silicon and stainless steel.
c. The maximum pressure in the measuring system can not exceed the rated overload value.
d. It is strictly forbidden to put any hard or sharp object into the pressure outlet hole.
e. The product is precision measuring instrument, it is forbidden to break at random, powerful clamping, disassemble or poke the lead hole with sharp metal implements. Please do not touch the flush diaphragm.
f. If the output is unusual, stop to check, if it is quality problems,return the products to the factory for repairing or replacement.
g. Please do not disassemble the circuit board of this instrument or change the other device by non-professional.
|SPECIFICATIONS FOR MRC-LT102 PRESSURE GAUGE|
|STATIC PRESSURE RANGE||Sensor|
|PROTECTION LEVEL/IP RATING||IP68|
|MEDIA||used in varied level measuring in petroleum, chemical engineering, metallurgy, electric power, pharmacy, water supply and environment protection.|
|SELECTIONS FOR MRC-LT-102 PRESSURE GAUGE|
|Measuring Range||Please contact us for customization|